Gauge is used to gauge two rail between the measurement of railway line, is a special measuring instrument
Rail Profile Gauge also known as route scan, digital track gauge, is used to measure gauge between the two tracks, is a special measuring instruments. Rail gauge ruler is characterized by ruler rod by the wooden rod and metal rod through the folding mechanism, the two rulers to form a composite rod ruler.
Digital Rail Gauge is a narrow gauge track detection device. Its structure is composed of a ruler body, scale and a swim frame on ruler body, the rule body is duplex profile structure, a square pipe shape, in the upper and lower profiles on both sides with the outward extension, in the inner edge of outer edge with a notch, insert stainless steel scale into the notch, then fixed it with pin on the end, ruler minddle install swim frame , side feet, side panels is fixed on the swim frame, side plate is fixed with adjusting bolts, ruler fitted with Cross plate and probe, the rail gauge was T-shaped structure.
Simple structure, easy production and installation, Digital Rail Gauge also has the advantages of high precision, durable, scale is not easy to wear, collapsible, light weight, easy to carry, reliable insulation, easy to maintain and use.
Aluminum rail gauge ruler with a carrying area of 42 × 25 × 2mm (wall thickness) aluminum alloy rectangular pipe, on the bottom of aluminum pipe have two 3.5mm stiffeners.
Wooden rail gauge tool with high quality white pine dry shape treatment, light weight, cheap.
Aluminum alloy rail seat, steel probe, good wear resistance.
Narrow rail gauge with area of 35 x 25 x 3mm aluminum alloy I-shaped profile.
Narrow rail gauge feet with rail wear-resistant nylon 66, steel probe have good wear resistance.
Aluminum rail gauge ruler with high-pressure polyethylene insulated jacket at both ends, the fixed end use ABS insulation splint, tube with insulation barrier, insulation performance is good.
When using rail profile gauge must adjust the roulette, alignment bubble zero level, look at the high scale, and then from the top of magnifying glass engraved line to see rail gauge scale.
It is forbidden to pull hard to avoid touching the two measuring heads of the gauge to reduce the error caused by the measurement.
Take good care of the dial on the rail profile gauge, it cannot be used once damaged.
After use promptly turned over to custody, scratched the stains on rail gauge ruler in time.
History Feedback: (0)